Stainless steel threaded bolt or internal thread
Self-locking nut (for threaded type)
Tmac® Cable Core Identifier
Clearly identifies the cores and their sequence in a three-phase power cable. Indication of the action to be taken to correct the core sequence, or to remove a fault on any cable core(s). Uses a transmitter unit with LED indicators at the measuring end, and an array of colour-coded diodes with connecting clips to fix the diode leads to the remote end of the cable to be checked.
The CCI is for use on DE-ENERGISED CABLES ONLY. Therefore correct operating procedures must be employed to ensure the cable is de-energised and any voltage discharged before connection of the CCI or its diode array.
- Two phases reversed
- Phases rotated forwards / backwards in sequence
- Earths in place on cable
- Equipment connected to cable (not properly isolated)
- Open-circuited or short-circuited cores